Product Name: Beijing Sincerity DMF-V-Series Micro Bend Coriolis Mass Flow Meters
Suitable for Liquid, Gas, Liquid- Solid, Gas- Solid, Liquid- Gas Mass Measurement or volume measurement
Material of the Measurement Pipeline
SS 316L/ HC Hastelloy
Please refer to the chart shown below. Special orders would be placed for high pressure.
-50℃~+150℃:-50℃~250℃;-50℃~ 350℃;-100℃~ 350℃
Flow Rate Measurement Accuracy
± 0.2% ,± 0.1% optional
Density Measurement Accuracy
±0.10% Flow Rate ±[½(Zero Point Stability/Flow Rate)×100]% Flow Rate
4~20mA Load Resistance <500Ω(Instantaneous or Density optional);
0~10kHz Instantaneous Flow Rate Pulse Signal;
Standard RS 485 RTU protocol
Explosion- Proof Symbol
EX d ib II C T5 Gb
Module Selection of the DMF-V-Series Coriolis Mass Flow Meters
Measurement scope (t/h)
Work Pressure (MPa)
DMF-V-Series Coriolis Mass Flow Meters (Micro-bend Promass) with core processor
DN 250 DMF-V-Series Coriolis Mass Flow Meters
(Micro-bend Promass) with core processor
Typical cased of different fields:
1.DN 250 DMF-V-Seires Coriolis Mass Flow Meter under Calibration
2. CNG Used in Cars:
3. Chemical Field:
4.Uncart. Oil Depot:
5. Used in Ships:
6. Cases in Bangladesh:
Beijing Sincerity Automatic Equipment Co,. Ltd is the manufacturer of producing Coriolis Mass Flow Meters. Our company was reformed from the China Petroleum &Chemical Corporation's Beijing Research Institute of Chemical Industries in 2006.
We have particular professional formula to produce the measuring pipeline compared to the other producing factories. Zeropoint Stability is good. Medium with tiny flow rate could be measured precisely, which reaches a minimum of 0- 1kg/h. Gas with either low pressure or with high pressure could be measured as well. Give us your problems, and you will never be disappointed!
Q: Why choosing Coriolis Mass Flow Meters but not the other flow meters?
Coriolis mass flow meters has huge difference with the volumetric flow meters.
All the other flow meters except the Coriolis mass flow meters could be called as volumetric flow meters, including, turbine flow meters, thermal mass flow meters, vortex flow meters, electromagnetic flow meters, etc.
For the Coriolis Mass Flow meters:
The measurement of the Coriolis Mass Flow meters will NOT be influenced by the factors of temperature, pressure, density, viscosity, etc. And the compensation calculation is NOT required.
The accuracy of the Coriolis Mass Flow Meters is ±0.1%, ±0.2% . The accuracy will NOT be influenced by the factors mentioned above.
Compared to this,
For the Volumetric Flow Meters:
The measurement of the volumetric flow meters WILL BE influenced by the factors of temperature, pressure, density, viscosity, etc. And the compensation calculation IS REQUIRED.
The accuracy of the volumetric flow meters is said to be reached by ±0.5%. While, the actual accuracy is hard to reach this level of ±1%. The accuracy will be influenced by factors as mentioned above.